How to use pressure steam sterilizer correctly
Last Update: 2014-09-02
Search more information of high quality chemicals, good prices and reliable suppliers, visit
1 Remove the outer packaging 2 Place the sterilizer correctly In order to make the sterilization work normally, the sterilizer must be placed on a solid horizontal surface 3 Add 3.5L water into the sterilizer, and add appropriate amount of water after each application to avoid burning the electric heating pipe due to water shortage 4 Put the prepared articles to be sterilized into the sterilizer, and leave a gap between the packages to ensure air penetration Cover the well and tighten the screws 5 Check the type of power supply and make sure it is consistent with the specification to be used Then plug one end into the plug of the sterilizer and the other end into the power socket 6 Observe the pressure indication or temperature indication of the pressure gauge When the steam pressure rises to about 0.165mpa, the safety valve will work automatically When the pressure falls to about 0.142mpa, the safety valve will close automatically to maintain constant pressure 7 After sterilization, unplug the plug and stop heating Open the vent again to release the steam Open the cover to discharge the remaining steam only after the pressure gauge pointer returns to zero.
This article is an English version of an article which is originally in the Chinese language on echemi.com and is provided for information purposes only.
This website makes no representation or warranty of any kind, either expressed or implied, as to the accuracy, completeness ownership or reliability of
the article or any translations thereof. If you have any concerns or complaints relating to the article, please send an email, providing a detailed
description of the concern or complaint, to firstname.lastname@example.org
. A staff member will contact you within 5 working days. Once verified, infringing content
will be removed immediately.